The MegaPulse DF-1P Fig. B.2 Current Carrying Tester is designed to generate the waveform described in AAMI DF-1, Figure B.2., for the connector cavity current-carrying test in AAMI DF-1, Annex B.
The test can be set to automatically conduct the first 490 cycles (500 ± 50 as shown in Standard) of the test, and then stop testing to allow laboratory personnel monitor the last 10 cycles in accordance with the Standard. Voltage and current BNC outputs make waveform analysis convenient.
Please take a look at the output voltage and current waveforms in the Manuals section below.
TestMinder USB control is included to allow automated repetitive testing.
A front-panel switch reverses the output polarity. Additionally provided with BNC oscilloscope output for voltage (100:1) and current (1V/A) for output monitoring.
The waveform output is 1.05uSec rise time, with a 18mS duration, with an output impedance of 15 ohms.
MegaPulse DF-1P Current carrying Tester ships complete, ready to perform the DF-1 test:
- TestMinder included to conduct automated repetitive testing using a Windows 32 bit Operating System
- BNC Current Jack (1V/ma)
- BNC Voltage jack (100:1)
- Graphs of theoretical and actual waveforms
- Calibration Certificate
- One-year warranty