For IEC Testing of residual voltage after plug disconnect. The CDT-240 disconnects the line at the voltage peak and analyzes results to the requirements of various Standards, allowing quick results. Scope output for viewing decay waveform. Presets for EN 61010, EN 60065, EN 60950, EN 60601-1 and EN60335. Recommendations for Pluggable Type B equipment. NIST calibration supplied; A2LA calibration available. 15A option for testing large equipment.
Questions? Contact us.
Minimum/Maximum Test Voltage: 90Vrms min/ 270Vrms max.
Maximum Test Current: 8A.
15A for testers with 15A option.
Fuse: 10A, 250V, time delay, 1-¼” x ¼”.
20A, 250V, time delay, 1-¼” x ¼” for testers with 15A option.
Probe Impedance: > 100MΩ, < 25pF
AC Peak Detection Accuracy: EUT disconnect within 5% of supply voltage peak.
Test Time Range: 0.50 – 10.0 Seconds.
Voltage Setting Range/Meas. Range: ±30 to 388V.
Voltage Setting Resolution: 1V.
Test Time Accuracy: ±1% for times ≥ 1 second, +/- 10ms for < 1 second.
Voltage Meter Accuracy: ±1.5% of instrument measuring range; ±2% ±0.8V up to 381Vpeak.
(See Appendix E for accuracy information with respect to voltage readings).
Rated Input Voltage: 100-240V.
Fuse: 2A, 250V, time delay, 5x20mm.
Safety Interlock: 2-pin terminal block on rear panel; shorted for operation.
BNC 100:1: Referenced to ground, 100:1 of measured voltage. (L-N, L-G, N-G) +/-2%.
L, N, G: Socket 4mm 1000V CAT III 32A.
Output receptacle options: NEMA 15A/125V, UK 12A/250V, SCHUCKO 16A/250V, CHINESE 10A/250V.