The MegaPulse DF-1P Fig. B.2 Current Carrying Tester is designed to generate the waveform
described in AAMI DF-1, Figure B.2., for the connector cavity
current-carrying test in AAMI DF-1, Annex B.
The test can be set to
automatically conduct the first 490 cycles (500 ± 50 as shown in Standard) of the test, and then stop testing to allow laboratory personnel
monitor the last 10 cycles in accordance with the Standard. Voltage and
current BNC outputs make waveform analysis convenient.
Please take a look at the output voltage and current waveforms in the Manuals section below.
TestMinder USB control is included to allow automated repetitive testing.
A front-panel switch reverses the output polarity. Additionally
provided with BNC oscilloscope output for voltage (100:1) and current
(1V/A) for output monitoring.
The waveform output is 1.05uSec rise time, with a 18mS duration, with an output impedance of 15 ohms.
MegaPulse DF-1P Current carrying Tester ships complete, ready to perform the DF-1 test:
TestMinder included to conduct automated repetitive testing using a Windows 32 bit Operating System
BNC Current Jack (1V/ma)
BNC Voltage jack (100:1)
Graphs of theoretical and actual waveforms
Write a Review
Generates the waveform required for the Current carrying Test, AAMI DF-1, Appendix B, and described in AAMI DF-1, Figure B.2.
Designed with operator safety in mind: Set up the EUT to the right and behind the tester, and a right-handed operator will find himself standing to the left of the tester, allowing maximum separation.
Theoretical graphs and actual graphs showing the output of your tester are included with each MegaPulse DF-1P, allowing easy sanity checks and proof to visiting agency engineers of test setups.
BNC Oscilloscope outputs allows quick connection and easy evaluation of voltage and current output waveforms.
Reverse polarity waveforms available by pushing a button on the front panel. No need to change the test setup. Positive polarity indication minimizes errors.
TestMinder computer control allows automation of the required 490 cycles; then the operator can conduct the last 10 cycles in which the current waveform must be monitored. (Waveform capture is not provided).
1-year calibration cycle.
All test leads included.
1 year warranty
Maximum Voltage output
Relativity humidity range
0-90 % Non condensing
AAMI DF-1 App B, Figure B.2
17 w x 17 L x 6 H
Continuos Full voltage
Power supply voltage
120 VAC 50/60 Hz
What is in the box?
Megapulse DF-1P Current carrying,2 high voltage Test leads,18 AWG Line power cord,Instruction manual
The HT-3500S RF tester is specifically designed for the challenges common to operating in the field. Along with heavy duty internal brackets and a sturdy, shock-absorbing case, this unit includes adjustable voltage output, voltage ramp, test time, leakage current and ground continuity limit settings - all controlled from an easy-to-access front panel. Call for pricing.
Perform one of the most difficult tests required with both speed and accuracy. The CDT-240 now offers a 16A option along with designed presets to implement the plug discharge test as described in IEC 61010, 60065 and 60335.
Compliance West USA is announcing the EOL (End-of-LIfe) of our MegaPulse Defib 5P series effective 07/14/2015. This unit has been replaced by our MegaPulse D5-PF advanced tester. Compliance West will continue to support, repair and calibrate the Defib 5P throughout the terms of any existing warranty agreements or up until the product's End-of-Support date. Please contact us with any questions.